Using the MENU Key R-1307
The Menu Key provides access to the operational functions of the R-1307, such as display mode, measurement units, dampening, display resolution and display brightness. In addition, communication parameters between the target and readout are set through the Menu Key.
To use the Menu Key, press and hold the key for 2 seconds to enter Configuration Mode, and then press the Menu Key to cycle through the menu selections. Press the UP arrow key (-) or the DOWN arrow key (¯) to change the settings for a selection.
|Menu Selection||Selectable Options||Meaning|
|Relative (datum) position Absolute (default) position|
|||||Dampening Number of Samples to Average|
|ID of first wireless scanning target
ID of second wireless scanning target
Note: For fixed beam wireless target or local target, set both upper and lower ID to the same number
|||0-9||System ID of Wireless Target(s) or local target|
|||1 (dim) to 5 (bright)||Display brightness|
Display position of target connected to local port
Request and display position of wireless target(s)
Display position of wireless targets controlled by another R-1307 or radio transceiver in Poll (Master) mode
|Following is an example of the information displayed for the selected
|Displays the PSD type (SC-100D or DL-10) target number, the laser mode (Pulsed or Fixed) and the PSD size (4×4 mm, 10×10 mm and 20×20 mm)|
Hidden menu items for any R-1307 model can be temporarily enabled by pressing and holding the UP arrow key while pressing the MENU key.
To exit configuration mode, press and hold the MENU button for approximately three seconds until the display returns to normal mode. The R-1307 will also return to normal mode automatically after approximately four seconds of inactivity.